Biomed Imaging Interv J 2007; 3(1):e17
© 2007 Biomedical Imaging and
Tomotherapy as a tool in image-guided radiation therapy (IGRT): current clinical experience and outcomes
S Yartsev1,*, PhD,
T Kron2, PhD, FCCPM,
J Van Dyk1,3, MSc, FCCPM
1 Physics and Engineering, London Regional
Cancer Program, London Health Sciences Centre, London, Ontario, Canada
2 Department of Physical Sciences, Peter MacCallum Cancer Centre,
East Melbourne, Australia
3 Departments of Oncology and Medical Biophysics, University of
Western Ontario, London, Ontario, Canada
Modern radiotherapy is characterised by a better target
definition through medical imaging accompanied by significantly improved
radiation delivery methods, most notably Intensity-Modulate Radiation Therapy
(IMRT). However, the treatment can only be as accurate as the positioning of
patients for their daily radiotherapy fraction. It is in this context that a
number of imaging modalities - ranging from ultrasound to on-board kilovoltage
imaging and computed tomography (CT) - have found their way into the treatment
room where they verify accurate patient positioning prior to or even during
delivery of radiation. Helical tomotherapy (HT) combines IMRT delivery with
in-built image guidance using megavoltage CT scanning. This paper discusses the
initial experience of different centres with IGRT using HT illustrated by a
number of clinical examples from the installation in London in Ontario, Canada,
one of the world�s first HT sites. We found that HT allows the delivery of
highly conformal radiation dose distributions combined with adequate daily
image acquisition. An important feature of this unit is its seamless
integration, which also includes a customised inverse treatment planning system
and a quality assurance module for individual patients. � 2007 Biomedical
Imaging and Intervention Journal. All rights reserved.
Keywords: Image guidance, helical tomotherapy, radiation
������� Clinical experience is always a crucial component in
the evaluation of any new technology in radiation therapy and helical
tomotherapy (HT) is no exception. In the early implementation stages, the
attention was focused on retrospective comparisons [1-9] of treatment plans
developed for different radiation delivery options in search of clinical
scenarios where HT is able to offer a significant improvement due to its
specific technological design, as discussed in the preceding review . These
plan comparisons concluded that indeed HT can provide improved normal tissue
sparing and highly conformal target coverage. Image-guided radiation therapy
(IGRT) available on HT, thanks to on-board megavoltage computed tomography
(MVCT) implemented in the commercially available Hi-ART model, allows daily
patient setup verification and repositioning. In this report, the first results
on its use in phantom studies and clinical practice are reviewed.
Clinical examples of image guidance in tomotherapy
MVCT was found to be an important imaging tool for precise
radiation delivery because it provides considerably more anatomical detail than
conventional radiation therapy port films used for patient setup verification.
There is a growing number of publications comparing treatment plans of
different radiotherapy techniques to HT delivery. The latter is predicted to
have some advantages, especially concerning homogeneity of the dose
distribution in the target [4,11-15]. In the following, we would like to
illustrate the specific characteristics of HT treatment plans using clinical
examples from our practice in London, in Ontario, Canada. Lung cancer was
chosen since the treatment outcomes are quite bad and there is an indication
that dose escalation may improve clinical outcomes . In order to achieve
this, the dose to normal lung must be reduced. This is a significant challenge
in radiation dose delivery. Head and neck, and prostate, the two other examples
chosen, are the most common applications for IMRT. Radiotherapy is often the
primary treatment modality in these diseases and in both cases, it has been
shown that normal tissue toxicity can be reduced by using advanced radiotherapy
Planning studies on the use of HT for treating localised lung
cancer showed more conformal dose distribution for the target and better
sparing of normal structures [2,17,18]. It would result in a better clinical
outcome for radiation treatment if the patient setup, target shape, size and
location remain the same as at the time when this patient was imaged for
planning. If these conditions are not met, a more conformal dose distribution
might partially miss the target and deliver high dose to the sensitive organs.
Several imaging techniques have demonstrated significant variations of the
tumour volume during radiotherapy treatment: electronic portal imaging showed
tumour shrinkage of 20% or more in 40% of patients , repeat kilovoltage CT
studies for 40 patients revealed a time trend towards decreasing gross tumour
volumes (GTVs) during fractionated stereotactic radiotherapy , and MVCT on
tomotherapy system allowed daily volumetric evaluation . These findings
indicate that periodic adjustments of treatment plans during a treatment course
are needed to account for changes in shape and location of the target volume
and critical structures when highly conformal techniques such as IMRT are used.
A pilot feasibility trial of 10 patients with non-small-cell lung cancer
provided results on contouring targets on tomotherapy MVCT and conventional CT
images . The volumetric agreement between conventional CT and MVCT was
excellent in 5 out of 7 patients with lesions located primarily in the lung
parenchyma while it was suboptimal for primary mediastinal disease. Kupelian et
al.  reported their study of tumour regression during external beam
radiotherapy for 10 patients with non-small-cell lung cancer. This tumour
reduction study using on-board MVCT on tomotherapy system gave full volumetric
evaluation, which was not possible by observations made on portal images
obtained during the course of treatment . MVCT scans of the targeted areas
were performed multiple times during treatment. The frequency of scanning was
determined by the treating physicians so that a total of 274 MVCT scans were
obtained on the 10 patients in the range of 9 to 35 scans per patient. Tumour
volumes were determined within the treatment planning system, and not by any
manual method. For all 10 tumours, the average decrease in volume was 1.2% per
day with a range of 0.6% to 2.3% per day. The lowest rate of shrinkage was
observed for the smallest lesion with an initial volume of 5.9 cm3.
The highest rate was observed in the largest lesion with an initial volume of
737 cm3. Other factors such as histology, level of necrosis and
dose-per-fraction may play a role in tumour size reduction . Direct
evaluation of tumour regression using MVCT immediately before treatment
indicates a potential necessity for plan updating during the treatment . If
tumour shrinkage during radiation treatment is clinically significant,
treatment plan re-optimisation should be considered, so that the dose to the
target remains as initially prescribed and improved sparing of sensitive
structures (such as normal lung volume) can be achieved . The clinical
significance of �plan updating� (or adaptive radiotherapy) remains to be
demonstrated and the correlation of tumour regression with clinical outcomes
should be studied . Another retrospective study of tumour regression during
treatment on a HT unit of 25 patients with lung cancer showed partial response
in 3 (12%), marginal response in 5 (20%) and stable disease in 17 (68%)
patients . Tumour regression of more than 25% was observed in 10 patients
(40%). However, the authors questioned the clinical significance of this
regression and field reductions during radiotherapy because there was no way to
document histological disease clearance. In our opinion, a follow-up for a
sufficiently long period of time after treatment may be a way of answering this
There are some cases of dramatic anatomy changes revealed by
MVCT imaging where a re-planning is absolutely necessary. Figure 1 demonstrates
one such example of a patient with non-small-cell lung cancer treated on
tomotherapy unit at our centre in 2005. Figure 1a shows one axial slice of the
kVCT study used for the initial treatment plan #1. However, when the patient
came for treatment 22 days after this study, the MVCT indicated (see Figure 1b)
that the tumour had displaced both anteriorly and superiorly probably as a
result of a collapse of one of the lung segments. Another kVCT study was
performed (see Figure 1c), which confirmed the findings on MVCT and was used
for creating plan #2. This plan was delivered for 18 fractions and daily MVCT
images showed only slight reduction of the GTV until the day when the
atelectasis was resolved and the tumour moved back as shown in Figure 1d. Then
yet another kVCT study was performed as shown in Figure 1e; it was used for
plan #3. The latter was applied for the remaining fractions and the MVCT image
on the final day of treatment is shown in Figure 1f. Clearly, such dramatic
changes in tumour/normal tissue anatomy can only be detected by daily CT
imaging of the patient.
Head & neck
Among the tumour sites, head and neck represent specific
challenges and opportunities for high-precision radiotherapy planning and
delivery. The proximity � in most cases � between the clinically manifest GTV
and the critical organs at risk and the fact that internal motion of tissues
and organs tends to be less of an issue in the head and neck region favors the
use of high precision IMRT . Hansen et al. have found that repeat CT
imaging and re-planning during the course of IMRT for patients with head and
neck cancer is essential because the clinical target volume decreased at a
median rate of 1.7%-1.8% per treatment day and the volume loss was frequently
asymmetric . Figure 2 illustrates the importance of daily setup corrections
in the case of the patient shown in Figure 3 from our companion paper . A
dose of 60 Gy to 90% of the planning target volume (PTV) was prescribed with
priority of sparing spinal cord and trachea. This patient lost weight during
treatment (from 133 kg at the start to 124 kg at the end of treatment),
resulting in changed patient anatomy especially in the treatment area. The
vector shifts used for daily patient setup alignment for this patient as a
function of time are shown in Figure 3.
Re-treatment of spinal metastasis is extremely difficult
because the spinal cord typically receives a radiobiological equivalent dose of
40 to 45 Gy given in 2 Gy fractions during the first course of treatment, so
that the risk of radiation myelitis after a second course is high . Mahan et
al. evaluated a feasibility of image-guided tomotherapy for re-treatment of
the vertebral spine . They performed measurements of dose gradients and
maximum cord doses using a cylindrical phantom, tested the ability of MVCT
images to localize spinal anatomy and used this experience for re-treatment of
8 patients with cord compressions to a mean dose of 28 Gy using HT. The total
imaging system error was measured by repeat imaging of an anthropomorphic head
phantom. At first, kVCT images were acquired on a CT simulator and transferred
to the tomotherapy database for image fusion. The phantom was placed in the
correct position relative to the machine isocenter, so that any non-zero setup
shifts calculated by the image guidance system represent error in the imaging
and fusion process. MVCT images were acquired over a 15 cm range superior-inferior and different options of the automatic image fusion algorithm were used. Total imaging system errors (1 ?) of ?0.6 mm, ?0.5 mm, and ?0.6 mm were obtained by "Bone", "Bone and Soft Tissue", and "Full-image" options, respectively. The uncertainty in the superior-inferior direction (?0.6 mm) was twice the uncertainty in the anterior-posterior and lateral directions (?0.3 mm).
It should be noted that the total clinical error may be much
larger because real patients can move during treatment and/or the vertebral
column can align in a different position relative to the treatment plan. In
radiotherapy of 8 patients with previously treated vertebral metastasis, Mahan et
al. acquired MVCT images through the PTV, autofused them with the planning
kVCT images, displaced the patients according to the calculated MVCT/kVCT shifts
and again performed MVCT study to verify that the shifts were correctly applied
and to assess intrafraction motion before treatment delivery. The range of the
total interfraction displacement with respect to the positioning on external
laser marks was as great as 15 mm . The standard deviation was 4.0 mm,
4.1 mm, and 4.3 mm in the anterior-posterior, lateral and superior-inferior
directions, respectively. Dose gradients of 10% per mm were found achievable by
HT in phantom measurements for a geometry representing the spinal cord as a 10
mm diameter cylinder and a 25 mm thick �vertebrae�. A very small 3 mm margin
was used for expanding the GTV to the PTV assuming high accuracy of positioning
and high dose gradients. As a result of such treatment 6 out of 8 patients had
complete relief of their pre-treatment symptoms, 2 had partial relief, and 2
died of distant disease during the mean follow-up of 15.2 months .
A group from Milan reported their estimates of systematic
and random set-up errors by using on-line as well as off-line setup correction
protocols. They have modeled the average systematic error and the residual
error for the case of post-operative prostate cancer (hypofractionated schedule
20 Gy in 2.9 Gy/fraction) and the minimum number of treatment sessions
necessary to correctly estimate systematic set-up error . Another study of
effectiveness of daily prostate registration on a tomotherapy unit was done by
K. Langen et al.  using 120 alignments from 3 patients with
implanted fiducial markers. They retrospectively compared manual registration
(i.e., visual matching of the prostate kVCT image used for planning and current
MVCT image) by different techniques. The reference alignment was calculated
based on the fiducial markers� centre of mass. If three implanted solid gold
markers clearly visible both in kVCT and MVCT were used, the relative number of
alignments, which differed by more than 3 mm from the reference, were 3%, 6%
and 3% in the anterior-posterior, superior-inferior and lateral directions,
respectively. If gold markers were disregarded and registration was done for
the prostate as seen on the MVCT with the prostate as seen on the planning kVCT
scan, the respective values for the same deviations were 24%, 33% and 3%. No alignments
differed more than 5 mm by this anatomy-based technique. If the organ contours
from the plan (the kVCT images as such were not used in this case) were used
for registration with the prostate MVCT image, there were more deviations from
the reference case by more than 3 mm: the respective values were 55%, 48% and
21%. The anatomy-based registrations outperformed the contour-based
registration both in terms of agreement with a marker-based centre-of-mass
reference alignment and inter-user variability.
The prostate gland can be identified with sufficient
contrast on MVCT images as shown for the patient treated in our institution on
images taken before and after registration with diagnostic kVCT study (Figure
4). Daily registration shifts for this patient are shown in Figure 5 for
anterior/posterior, lateral and superior / inferior directions. The couches
(�patient support assembly�) on diagnostic CT and tomotherapy unit have
different mechanical properties, so a heavier patient with the same initial setup
made on external marks is in a lower position on the more flexible couch on
tomotherapy. This is detected, taken into account and corrected by MVCT
Song et al. evaluated the image-guidance capabilities
of MVCT by comparing volumes of prostate contours outlined on MVCT and kVCT
studies of the same 5 patients . Seven observers did the contouring twice
with an interval of 2 months, which allowed the evaluation of inter- and
intraobserver variability. The volumes defined on the kVCT were smaller and more consistent compared with the MVCT results (54.1 ? 8.6rms cm3 and , 59.9 ? 14.8rms cm3, respectively). On average, the increase in the clinical target volume variability ?? = ?(?2MVCT - ?2kVCT), for both interobserver and intraobserver studies was 0.32 cm .
This outcome is not surprising, because observers tend to segment larger volumes on images with lower soft-tissue contrast [34,35]. Song et al. suggested that the techniques that do not require contours for deformable-image registration, such as intensity-driven dose-warping techniques [36,37] may be more suitable for MVCT.
When imaging the thorax or abdomen of a patient,
respiration-induced artefacts such as blurring, doubling, streaking and
distortion degrade the image quality and affect the target localisation ability
. These artefacts depend on the ratio between breathing cycle and the
gantry rotation speed. In conventional CT or a modern CT scanner, each rotation
of the scan can be completed within 1 s or less, during which the organ motion
is relatively small. A single slice of MVCT in HT is reconstructed from a 180o
rotation in 5 s, which will introduce motion artefacts from breathing. However,
other competing techniques suffer from the same problem, e.g., a cone-beam CT
scan takes typically 45 s to 1 min for acquiring the projection data in a full
360o scan . Some investigations of motion artefacts in
tomotherapy imaging were done [40-42] and it can be expected that future
developments will allow for faster image acquisition. Several groups have also
worked on gated HT delivery [43,44]. It will be interesting to see how similar
approaches can be utilised for improved MVCT acquisition in the future.
Clinical experience with HT is rapidly growing, stimulated
by encouraging dosimetric results from planning studies of this method in comparison
with traditional techniques. Preliminary results of implementation of IGRT in a
tomotherapy setting shows that the on-board MVCT image acquisition system
allows improved patient positioning. Increased setup precision permits the use
of smaller margins around targets and organs at risk. Clinical experience in
different institutions has proved the usefulness of MVCT imaging for
corrections of patient setup leading to the possibility of better tumour
control and a better sparing of healthy tissues. IGRT benefits individual
patients and also, by combining information from many patients, allows
radiotherapy departments to develop rational strategies for margin design and
the identification of potential weaknesses in the treatment chain.
This study was conducted with the support of the Ontario Institute for Cancer Research through funding provided by the government of Ontario.
Figure 1 Screenshots of images of the patient with lung cancer for (a) initial kVCT image done 35 days before the treatment start, (b) MVCT image taken 13 days before the treatment start, (c) second MVCT image done 4 days before the treatment start, (d) MVCT image taken before fraction 18 on day 26 of the treatment, (e) third kVCT image done on day 33 of the treatment, (f) MVCT image taken on the last (52nd) day of the treatment, fraction 30.
Figure 2 kVCT images (left) before the treatment and (right) after delivery of 22 fractions for the patient with significant weight loss.
Figure 3 The values of vector shifts introduced daily based on matching of the PTV contour used for planning (red line on Fig. 2a) with the daily MVCT images.
Figure 4 Image showing both kVCT (grey) and MVCT (green) images taken (top) before and (bottom) after registration.
Figure 5 Daily shifts in different directions after registration of the patient with prostate cancer. The adjustments in anterior/posterior direction have a systematic shift due to different mechanical properties of the couches in the diagnostic CT scanner and the tomotherapy unit.
Grigorov G, Kron T, Wong E et al. Optimization of helical tomotherapy treatment plans for prostate cancer. Phys Med Biol 2003; 48(13):1933-43.
Kron T, Grigorov G, Yu E et al. Planning evaluation of radiotherapy for complex lung cancer cases using helical tomotherapy. Phys Med Biol 2004; 49(16):3675-90.
van Vulpen M, Field C, Raaijmakers CP et al. Comparing step-and-shoot IMRT with dynamic helical tomotherapy IMRT plans for head-and-neck cancer. Int J Radiat Oncol Biol Phys 2005; 62(5):1535-9.
Yartsev S, Kron T, Cozzi L et al. Tomotherapy planning of small brain tumours. Radiother Oncol 2005; 74(1):49-52.
Han C, Liu A, Schultheiss TE et al. Dosimetric comparisons of helical tomotherapy treatment plans and step-and-shoot intensity-modulated radiosurgery treatment plans in intracranial stereotactic radiosurgery. Int J Radiat Oncol Biol Phys 2006; 65(2):608-16.
Sheng K, Molloy JA, Read PW. Intensity-modulated radiation therapy (IMRT) dosimetry of the head and neck: a comparison of treatment plans using linear accelerator-based IMRT and helical tomotherapy. Int J Radiat Oncol Biol Phys 2006; 65(3):917-23.
Fiorino C, Dell'Oca I, Pierelli A et al. Significant improvement in normal tissue sparing and target coverage for head and neck cancer by means of helical tomotherapy. Radiother Oncol 2006; 78(3):276-82.
Pezner RD, Liu A, Han C et al. Dosimetric comparison of helical tomotherapy treatment and step-and-shoot intensity-modulated radiotherapy of retroperitoneal sarcoma. Radiother Oncol 2006; 81(1):81-7.
Cozzi L, Clivio A, Bauman G et al. Comparison of advanced irradiation techniques with photons for benign intracranial tumours. Radiother Oncol 2006; 80(2):268-73.
Yartsev S, Kron T, Van Dyk J. Tomotherapy as a tool in image-guided radiation therapy (IGRT): theoretical and technological aspects. Biomed Imaging Interv J 2007; 3(1):e16.
Langen KM, Meeks SL, Poole DO et al. The use of megavoltage CT (MVCT) images for dose recomputations. Phys Med Biol 2005; 50(18):4259-76.
Han C, Liu A, Schultheiss TE et al. Dosimetric comparisons of helical tomotherapy treatment plans and step-and-shoot intensity-modulated radiosurgery treatment plans in intracranial stereotactic radiosurgery. Int J Radiat Oncol Biol Phys 2006; 65(2):608-16.
Penagaricano JA, Papanikolaou N, Yan Y et al. Feasibility of cranio-spinal axis radiation with the Hi-Art tomotherapy system. Radiother Oncol 2005; 76(1):72-8.
Hui SK, Kapatoes J, Fowler J et al. Feasibility study of helical tomotherapy for total body or total marrow irradiation. Med Phys 2005; 32(10):3214-24.
Wong JY, Liu A, Schultheiss T et al. Targeted total marrow irradiation using three-dimensional image-guided tomographic intensity-modulated radiation therapy: an alternative to standard total body irradiation. Biol Blood Marrow Transplant 2006; 12(3):306-15.
Kong FM, Hayman JA, Griffith KA et al. Final toxicity results of a radiation-dose escalation study in patients with non-small-cell lung cancer (NSCLC): predictors for radiation pneumonitis and fibrosis. Int J Radiat Oncol Biol Phys 2006; 65(4):1075-86.
Mehta M, Scrimger R, Mackie R et al. A new approach to dose escalation in non-small-cell lung cancer. Int J Radiat Oncol Biol Phys 2001; 49(1):23-33.
Scrimger RA, Tome WA, Olivera GH et al. Reduction in radiation dose to lung and other normal tissues using helical tomotherapy to treat lung cancer, in comparison to conventional field arrangements. Am J Clin Oncol 2003; 26(1):70-8.
Erridge SC, Seppenwoolde Y, Muller SH et al. Portal imaging to assess set-up errors, tumor motion and tumor shrinkage during conformal radiotherapy of non-small cell lung cancer. Radiother Oncol 2003; 66(1):75-85.
Underberg RW, Lagerwaard FJ, van Tinteren H et al. Time trends in target volumes for stage I non-small-cell lung cancer after stereotactic radiotherapy. Int J Radiat Oncol Biol Phys 2006; 64(4):1221-8.
Kupelian PA, Ramsey C, Meeks SL et al. Serial megavoltage CT imaging during external beam radiotherapy for non-small-cell lung cancer: observations on tumor regression during treatment. Int J Radiat Oncol Biol Phys 2005; 63(4):1024-8.
Welsh JS, Bradley K, Ruchala KJ et al. Megavoltage computed tomography imaging: a potential tool to guide and improve the delivery of thoracic radiation therapy. Clin Lung Cancer 2004; 5(5):303-6.
Dar AR, Yartsev S, Rodrigues G et al. Serial megavoltage CT imaging during external beam radiotherapy for non-small-cell lung cancer: in regard to Kupelian et al. (Int J Radiat Oncol Biol Phys 2005;630:1024-1028). Int J Radiat Oncol Biol Phys 2006; 64(1):328; author reply 328-9.
Ramsey CR, Langen KM, Kupelian PA et al. A technique for adaptive image-guided helical tomotherapy for lung cancer. Int J Radiat Oncol Biol Phys 2006; 64(4):1237-44.
Kupelian PA, Ramsey C, Meeks SL et al. In response to Dr Dar et al. Int J Radiat Oncol Biol Phys 2006; 64:328-9.
Siker ML, Tome WA, Mehta MP. Tumor volume changes on serial imaging with megavoltage CT for non-small-cell lung cancer during intensity-modulated radiotherapy: how reliable, consistent, and meaningful is the effect? Int J Radiat Oncol Biol Phys 2006; 66(1):135-41.
Bentzen SM. Radiation therapy: intensity modulated, image guided, biologically optimized and evidence based. Radiother Oncol 2005; 77(3):227-30.
Hansen EK, Bucci MK, Quivey JM et al. Repeat CT imaging and replanning during the course of IMRT for head-and-neck cancer. Int J Radiat Oncol Biol Phys 2006; 64(2):355-62.
Marcus RB Jr, Million RR. The incidence of myelitis after irradiation of the cervical spinal cord. Int J Radiat Oncol Biol Phys 1990; 19(1):3-8.
Mahan SL, Ramsey CR, Scaperoth DD et al. Evaluation of image-guided helical tomotherapy for the retreatment of spinal metastasis. Int J Radiat Oncol Biol Phys 2005; 63(5):1576-83.
Broggi S, Fiorino C, Cattaneo GM et al. MVCT - guided imaging. Radiother Oncol 2005; 76(Suppl 2):S7.
Langen KM, Zhang Y, Andrews RD et al. Initial experience with megavoltage (MV) CT guidance for daily prostate alignments. Int J Radiat Oncol Biol Phys 2005; 62(5):1517-24.
Song WY, Chiu B, Bauman GS et al. Prostate contouring uncertainty in megavoltage computed tomography images acquired with a helical tomotherapy unit during image-guided radiation therapy. Int J Radiat Oncol Biol Phys 2006; 65(2):595-607.
Fiorino C, Reni M, Bolognesi A et al. Intra- and inter-observer variability in contouring prostate and seminal vesicles: implications for conformal treatment planning. Radiother Oncol 1998; 47(3):285-92.
Roach M 3rd, Faillace-Akazawa P, Malfatti C et al. Prostate volumes defined by magnetic resonance imaging and computerized tomographic scans for three-dimensional conformal radiotherapy. Int J Radiat Oncol Biol Phys 1996; 35(5):1011-8.
Wang H, Dong L, O'Daniel J et al. Validation of an accelerated 'demons' algorithm for deformable image registration in radiation therapy. Phys Med Biol 2005; 50(12):2887-905.
Wang H, Dong L, Lii MF et al. Implementation and validation of a three-dimensional deformable registration algorithm for targeted prostate cancer radiotherapy. Int J Radiat Oncol Biol Phys 2005; 61(3):725-35.
Sonke JJ, Zijp L, Remeijer P et al. Respiratory correlated cone beam CT. Med Phys 2005; 32(4):1176-86.
Li T, Schreibmann E, Yang Y et al. Motion correction for improved target localization with on-board cone-beam computed tomography. Phys Med Biol 2006; 51(2):253-67.
Yang JN, Mackie TR, Reckwerdt P et al. An investigation of tomotherapy beam delivery. Med Phys 1997; 24(3):425-36.
Lu W, Mackie TR. Tomographic motion detection and correction directly in sinogram space. Phys Med Biol 2002; 47(8):1267-84.
Gagne IM, Robinson DM. The impact of tumor motion upon CT image integrity and target delineation. Med Phys 2004; 31(12):3378-92.
Zhang T. Respiratory gating and 4-D tomotherapy. Med Phys 2004; 31:3529.
Kim B, Kron T, Chen J et al. Feasibility of gated helical tomotherapy using a real time position management system. Med Phys 2005; 32:2428.
|Received 19 December 2006; accepted 30 January 2007
Correspondence: Department of Physics & Engineering, London Regional Cancer Program, London Health Sciences Centre, 790 Commissioners Road East, London, Ontario, Canada N6A 4L6. Tel.: +1-5196858600; Fax: +1-5196858658; E-mail: email@example.com (Slav Yartsev).
Please cite as: Yartsev S, Kron T, Van Dyk J,
Tomotherapy as a tool in image-guided radiation therapy (IGRT): current clinical experience and outcomes, Biomed Imaging Interv J 2007; 3(1):e17
This article has been viewed 5624 times.